A supplemental nursing system (SNS) allows a mother to feed her baby supplemental milk at the breast. It can be used as a short-term aid on the way back to full breastfeeding and can sometimes be used longer-term. The SNS feeding tube can also be used as a way to finger-feed your baby either as a sucking exercise or when transitioning your baby back to the breast. Keeping in touch with an IBCLC lactation consultant is advisable while you are using an SNS. For the pros and cons of this method of feeding and further information on how to use them correctly see Breastfeeding With a Supplemental Nursing System. This article describes how to make a homemade supplemental nursing system.
Making a Homemade Supplemental Nursing System
You will need
- An ordinary feeding or storage bottle and teat
- A 5 or 6 French, 50cm or 90cm feeding tube1
- A 5ml syringe for flushing and cleaning the tube—your local pharmacy may sell these or online stores like Amazon sell individually wrapped sterile syringes.
- Slightly enlarge the hole of a bottle teat or vent (with clean scissors) so that you can feed the tubing upwards through the teat. The plastic attachment (an open/close valve) on the feeding tube sits in the milk at the bottom of the bottle (make sure the valve is open). A tight fit for the tube through the bottle nipple will reduce spillage.
- Add your expressed breastmilk or formula supplement to the bottle. Screw the artificial teat onto the bottle with the supplemental tube threaded through it and the valve end of the tube in the milk.
- Piercing a second hole in the teat, or using a teat with a vent, allows air into the bottle so your baby doesn’t need to suck harder as the bottle empties. Some mothers find it works well without the extra hole.
- Place the bottle on a table close to where you are feeding your baby, or put it in a shirt pocket, between your breasts or between your knees depending on the length of the tube.
- The free rounded end of the tube should lie alongside mother’s nipple and be taken into the baby’s mouth when he begins to suck. Alternatively you can latch your baby on the breast first, and then slide the tube into the corner of the baby’s mouth when you are ready (for example after he has breastfed on both breasts first).
- Some mothers find it useful to tape the tube to their breast and others prefer not to, experiment to find what works for you. If you tape the tube to your breast before your baby latches, make sure the tape is far enough away from your nipple so that it won’t interfere with your baby’s latch. Also check that the tube isn’t extended further than your nipple or it may make your baby gag or reach the back of his throat (taping it to the breast will help avoid this). Jack Newman, a Canadian paediatrician and breastfeeding expert says the tube only needs to be over the baby’s gums to work properly (Newman, 2009)
Adjusting milk flow
If your baby can make a good seal on the the breast with the tube in place, you will soon see milk rise up the tube into his mouth and he will start swallowing. If the flow of milk seems too fast and your baby is gulping milk too quickly you can adjust the flow by;
- Adjusting the height of the bottle of milk. The bottle containing the supplement should not need to be held higher than your baby’s head to work. The higher the bottle the faster the flow of milk.
- Different gauge tubes will alter milk flow. The wider the diameter of the tube the faster the flow of milk, 6 French (6 fr) is a wider tube than 5 fr.
- Pinching the tube to narrow the diameter and slow the flow.
If your baby struggles to suck milk up the tube to get the flow started he may start to get frustrated. Filling the tube with breast milk using the 5 ml syringe will require less work and give him an instant reward. You could also experiment lifting the bottle higher to get the flow started.
Tips from Jack Newman
Jack Newman, a Canadian paediatrician and breastfeeding expert has some tips for using a lactation aid in the following article including the benefits of using one and how to wean your baby from using it when you are ready:
[The lactation aid] does seem to function better if the tube is placed in the corner of the baby’s mouth and enters straight into the baby’s mouth over the tongue. (Point it slightly to the roof of the baby’s mouth). It is occasionally helpful for the mother to hold the tube in place with her finger, as some babies tend to push the tube out of position with their tongues.
Making a Supplementary Nursing System, Canadian Childbearing Almanac
Cleaning the Homemade Supplemental Nursing System
You can clean the bottle and teat as usual. To clean the tubing; first push cold water through the tube with a 5ml syringe straight after use to remove any milk residue, then flush through again with hot soapy water. Finally rinse thoroughly and see Jack Newman’s instructions below. Jack Newman recommends against boiling the tubes in a home made supplementer but check the manufacturer’s instructions for a commercial supplementer.
Cleaning the Device
Do not boil the tube of the non-manufactured aid. It is not made to be boiled.
After using the device, clean the bottle and nipple as usual. Do not boil the tube. The tube should be emptied after use and then rinsed through with hot water (suck up hot water into the tube from a cup) and then hung up to dry. Soap, though not necessary, may be used if desired, but rinse the tube well. Tubes may become stiff and unsuitable for use after a few days to a week.
A homemade supplemental nursing system can be used as a way to give your baby supplemental milk at the breast or via finger feeding. It is an alternative to cup feeding, or using a bottle. However it is not suitable for all babies. A supplemental nursing system has a number of disadvantages as well as advantages and is best used under the supervision of an IBCLC lactation consultant or other health professional. This article should be read in conjunction with Breastfeeding With a Supplemental Nursing System.